MLZ is a cooperation between:> Technische Universität München> Helmholtz-Zentrum Hereon > Forschungszentrum Jülich
MLZ is a member of:
MLZ on social media:
Polarized hot neutron diffractometer
A third measurement method named flipping-ratio is under development and planned to be operational using a new 8 T magnet in 2016.
SNP allows distinguishing between the depolarisation and polarisation rotation occurring during the scattering in the sample. X, Y, Z components of the scattered polarisation are measured for each orientation of the incoming polarisation and hence a polarisation matrix of 9 elements for an individual Bragg reflection can be found. Determining the relationship between the directions of incident and scattered polarisations gives access to the 16 independent correlation functions involved in the most general nuclear and magnetic scattering processes. Generally, this leads to the direction’s determination of the magnetic interaction vector of magnetic structures. For those structures coinciding nuclear and magnetic reflections in reciprocal space, SNP leads to the amplitude’s determination of the magnetic interaction vectors, and hence to the magnetisation distribution. SNP could also be employed to study the magnetic domain distribution.
Actually, polarised 3He spin filters are used on POLI in order to produce and to analyse neutron polarisation. This method offers best performance in the polarising of hot neutrons. Tuning the pressure of 3He in the filter cell it is possible to optimise the polariser’s performance depending on wave length or experimental needs. The automatic correction of the time-dependent polarisation is applied. Because of the 3He filters’ sensitivity about non-homogeneous magnetic fields, a dedicated solid state supermirror bender will be used in the future for experiments with high magnetic fields.
Primary beam: SR-9a on hot source
Neutron polarisation: 3He spin filter
|with Cryopad||without Cryopad|
|-10° < 2θ < 120°||-30° < 2θ < 130°|
|-180° < ω < 180°||-180° < ω < 180°|
|-4° < ξ1 < 4°||-5° < ξ1 < 5°|
|-4° < ξ2 < 4°||-5° < ξ2 < 5°|
|v = 0||-4.2° < v < 30°|
Cryopad (zero-field polarimeter)
Dr. Vladimir Hutanu
Phone: +49 (0)89 289-12153
Dr. Hao Deng
Phone: +49 (0)89 289-54725
Phone: +49 (0)89 158860-518
Find the latest publications regarding POLI in our publication database iMPULSE:
Heinz Maier-Leibnitz Zentrum. (2015). POLI: Polarised hot neutron diffractometer. Journal of large-scale research facilities, 1, A16. http://dx.doi.org/10.17815/jlsrf-1-22
For citation please always include the DOI.